The only input commands I did were to set up teh channel definition and source setup. I have no experience with LAbView so far. However I get this error and nothing is showing up not on the equipment neither in the VI interface in LabView. I will give it a try. This is where I get error D 00 40 where after rading through the pages of the manual I got to the conclusion that D stands for data error, first two characters, xx, stand for X axis, the second set of 2 characters Y axis
|Date Added:||14 January 2014|
|File Size:||30.25 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Moreover I cannot see the trace of the sweep I expected a trace of 10 MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface.
The VI instrument driver works. The only input commands I did were to set up teh channel definition and source setup.
Agilent Technologies / Keysight Technologies hp4145 Analyzer
I will try nevertheless to find a solution. If i were you, i would contact KeySight formerly known as Agilent formerly known as HP what that message on the device is meaning. Message 4 of This is on the B, right?
Message 5 of However I get this error and nothing is showing up not on the equipment neither in the VI interface in LabView. I will give it a try. Did you try other commands which take a longer time to execute? Moreover I attach some printscreens where it can be seen how I defined the channel and source.
Next step I have tried before preparing the probestation was to set for one of the SMUs SMU1 a voltage value and measure the set value using a multimeter. I really appreciate all the feedback I can receive.
Solved: LabView VI for HP : diode measurement IV – Discussion Forums – National Instruments
Message 9 of What exactly is stopping us from doing this? Such error can occur if configuration of GPIB is not correct e.
During this time the multimeter which is connected to SMU1 output of the equipment register the voltage value. I will keep you posted with the outcome.
I looked on the manual for this error and I cound not find it. Message 6 of I am not sure if the Instrument Driver for the B adds custom messages. The data I want to represent is IV curves. I am new to the forum and also new to LabView.
Message 10 of Error 6 I get it only when pressing “Take Measurement” labvoew. This would suggest me that there is communication through the GPIB. Message 3 of labvlew Most Active Software Boards: I can now input data from Labview laabview HP The rest of the channels are not assigned, as now I only try one port measurement, to see if I can take out data from the HP This is where I get error D 00 40 where after rading through the pages of the manual I got to the conclusion that D stands for data error, first two characters, xx, stand for X axis, the second set of 2 characters Y axis I think there should be somewhere in the block diagrams of the HP B Example.
HP B Semiconductor Parameter Analyzer – Page 2 – HxC Floppy Drive Emulator
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. You can right click on the error cluster which shows error 6 and select “Explain Error”.
I would like to know labivew there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from the HP via LabView? Message 2 of